- This topic has 1 reply, 1 voice, and was last updated September 27, 2018 at 9:20 pm by .
- September 27, 2018 at 9:20 pm #391224
During the use of some TB – Temporary Bipolar Pacing Leads, featuring the 2mm unshrouded connectors, the connector cap housing (see Picture 1, No. 2 Pin Cap and Cover) may slide and potentially expose the connection wire. In some instances, this may cause the wire to be more susceptible to loss of connectivity or breakage during movement of the cables causing interruption of the pacing system.
The analysis of the returned devices attributed the failure to a design change of the cap housing of the pins. In the last six years, a total of four serious injuries were reported to Oscor which were attributed to a connector cap malfunction causing the lead connector to separate during use potentially leading to an interruption of the pacing system.
This notice provided by Budget101.com
- You must be logged in to reply to this topic.