Results 1 to 1 of 1
09-23-2013, 11:15 PM #1
- Join Date
- Aug 2012
- Merced, CA 95348-1422
- Blog Entries
Recall: Outlook Safety Infusion System Burette Set. For intravenous fluid
Product Description Outlook Safety Infusion System Burette Set. For intravenous fluid administration.
Recall Number Z-2166-2013
Classification Class II
Code Info Catelog Number 375038, Lot Number 0061317318, Expiration Date: 04/30/2016
Product Distributed Qty 900
Reason For Recall Reports were received of an incorrectly assembled Outlook
Safety Infusion System Burette Set (Product Code 375038), and the investigation determined that one of the ULTRASITE injection sites and corresponding low pressure backcheck valve was assembled upside down.
Event Id 65993
Product Type Devices
B. Braun Medical, Inc.
Voluntary / Mandated Voluntary: Firm Initiated
Recall Initiation Date 2013-08-05
Initial Firm Notification of Consignee or Public Letter
Distribution Pattern Distributed in the states of FL, NY, NJ, and TX.