Results 1 to 1 of 1
10-19-2012, 12:20 PM #1
Ventlab Corporation Issues Nationwide Recall of its Manual Resuscitators
On July 11, 2012, Ventlab Corporation initiated a nationwide recall of 14,602 of its manual resuscitators. The manual resuscitators as listed below have been found to potentially deliver little to no air/oxygen through the patient valve to the patient, which could result in life threatening health consequences that include hypoxia and hypoventilation.